AirSense™ 10 Elite
The ResMed AirSense 10 Elite is a premium fixed-pressure device with integrated humidifier, cellular connectivity and advanced event detection.
With each AirSense 10 device offering built-in cellular technology, you get unprecedented access to therapy data and remote access to device settings, helping you to be more connected with your patients and even more efficient in your business.
The AirSense 10 Elite includes central sleep apnea (CSA), respiratory effort related arousal (RERA) and Cheyne-Stokes respiration (CSR) detection to provide treating physicians valuable insight into the continued health of your patients.
The algorithm has been designed to provide data to assist the physician in identifying indications of CSA, RERA and CSR, which you can monitor through AirView, to quickly and accurately provide patients with information to seek the right treatment.
|AirSense 10 CPAP||AirSense 10 Elite||AirSense 10 AutoSet||AirSense 10 AutoSet for Her|
|AutoSet for Her|
|Central apnea detection|
|Cheyne-Stokes respiration detection|
|Expiratory pressure relief (EPRTM)|
|Built-in cellular technology|
|AirView's Remote Assist|
|Dimensions (H x W x D) (Device and water tub)||4.57" x 10.04" x 5.91"|
|Air filter||Standard: Average arrestance: > 75% for ~7 micron dust
||Hypoallergenic: Efficiency: >98% for ~7-8 micron dust. >80% for ~0.5 micron dust.
|Aircraft use||Meets Federal Aviation Administration requirements for all phases of air travel.|
|Operating pressure range||4 to 20 cm H2O|
|Pressure level measured according to ISO 17510-1:2007 (CPAP mode):|
|SlimLine or Standard air tubing||26.6 dBA with uncertainty of 2 dBA|
|Power level measured according to ISO 17510-1:2007 (CPAP mode):|
|SlimLine or Standard air tubing||34.6 dBA with uncertainty of 2 dBA|
|Sound (Declared dual-number noise emission values in accordance with ISO 4871:1996.)|
|Electromagnetic compatibility||Complies with all EMC requirements according to IEC60601-1-2:2007 for residential, commercial and light industry environments. It is recommended that mobile communication devices be kept at least 3’3” away.|