Designed to comfortably fit a broader range of patients, AirFit N20 features an entirely new cushion design. To make fitting your existing patients easier, we recommend using our sizing guide – included in each AirFit N20 box and specifically designed to help patients find the right fit.
You can also download a copy of it from the ResMed website.
The sizing guide has three cut-outs: small, medium and large. Find the size that accommodates your patient. The modular design of AirFit N20 means each size cushion easily clicks into the same mask frame for easier setup.
Unclip EasyClick Magnets.
With your patient holding the mask towards their face, slide the headgear over the back of their head.
Reclip EasyClick Magnets.
Adjust the tabs on the upper headgear straps.
Adjust the tabs on the lower headgear straps.
The upper headgear straps provide most of the adjustment. The lower headgear straps can be used to gently tighten a patient’s mask. Do not overtighten as this may cause discomfort.
If your patient experiences leaks, or for additional information on mask maintenance and care, you can also visit the Supporting your patients section for more videos.
AirView is ResMed’s cloud-based patient management system that provides remote access to your patients’ therapy data. Patients using AirSense™ 10 or AirCurve™ 10 devices with built-in wireless connectivity can be remotely monitored. * Wireless communication depends on network availability.
With the mask held against your face, pull the headgear over your head.
Bring each of the magnets up to meet their corresponding clips on the frame.
Adjust the tabs on the upper headgear.
Adjust the tabs on the lower headgear.
With your device still switched off, connect your device tubing to your mask elbow.
Connect the elbow to your mask.
Switch on your therapy device.
Managing leak with your AirFit N20
Leak often occurs when your mask’s seal is compromised by movement, incorrect fitting or improper sizing.
To find a source of leak, run your finger along the edge of your mask’s seal. Once located, adjust your headgear.
Resolve any leak in the upper part of your mask by adjusting your upper straps.
Resolve lower mask leak with your lower straps.
If leak persists, reseal by briefly pulling your mask away from your face and letting your cushion reinflate.
If leak continues, you may need a different size mask. Try our sizing guide or speak to your healthcare provider about getting a different size mask.
Cleaning your AirFit N20 (Daily)
Soak cushion, elbow and short tubing in warm water (approximately 86°F/30°C) with mild liquid detergent for up to ten minutes. To ensure a thorough clean, shake components in water.
Clean components with a soft-bristle brush.
Rinse components under warm water. Inspect each component to ensure that each is clean and free of detergent residue. Wash and rinse again, if necessary.
Leave mask components to air dry out of direct sunlight.
Cleaning your AirFit N20 (Weekly)
Disassemble mask. EasyClick Magnets can remain attached to headgear during cleaning.
Soak frame and headgear in warm water (approximately 86°F/30°C) with mild liquid detergent for up to ten minutes. To ensure a thorough clean, shake frame and headgear in water.
Rinse both components in clean water. Squeeze each repeatedly to ensure detergent is fully removed. Inspect each component to ensure that each is clean and free of detergent residue. Wash and rinse again, if necessary.
Squeeze each component repeatedly to remove excess water.
Leave both components to air dry out of direct sunlight.
How to Disassemble
Remove elbow from frame.
Unclip the EasyClick Magnets.
Pull strapping away from frame.
Hold the frame by placing your thumb over the side slot. Pull cushion away from frame.
How to Reassemble
Push the cushion into the frame. The frame has a shape that allows you to insert the cushion only one way.
With the ResMed logo on the headgear facing outside and upright, attach the EasyClick Magnets.
Thread the upper headgear straps into the frame slots from the inside. Fold them over to secure.