For additional therapy assistance.
For additional therapy assistance.
Recent studies have shown that, when comparing against the current market-leading full-face mask, 86% of patients prefer the seal of AirFit F20.1
In terms of stability and ease of use, the majority of patients studied also prefer AirFit F20 over the current market leader1, demonstrating that our technologies work to improve patient experiences. Through extensive studies, we have designed a Full-face mask to deliver superior therapy performance and a smoother, easier patient experience.
To comfortably fit a broader range of patients, AirFit F20 features an entirely new cushion design. To make crosswalking your existing patients easier, we recommend using our sizing guide – included in each AirFit F20 box and specifically designed to help patients find the right fit. You can also download a copy of our sizing guide here.
The arrow of your sizing guide should align with the top of your patient’s nasal bridge. The segment of the guide corresponding with the crease below your patient’s lower lip will indicate their best fit. If a patient falls between sizes, we recommend selecting a larger option for greater comfort.
AirFit F20 has three cushion sizes - small, medium and large. The modular design of the AirFit F20 means each size cushion easily clicks into the same mask frame.
Your patients can also download a copy of our sizing guide from the ResMed website.
Step 1 - Unclip the EasyClick Magnets.
Step 2 - With the mask held against your face, pull the headgear over your head. Ensure the top of the mask cushion is sitting comfortably at the top of your nasal bridge. The bottom of your headgear should settle comfortably at the top of your neck - with the top of the headgear sitting on your crown.
Step 3 - Bring each of the magnets up to meet their corresponding clips on the frame.
Step 4 - Adjust the tabs on the upper headgear.
Step 5 - Adjust the tabs on the lower headgear.
Step 6 - With your device still switched off, connect your device tubing to your mask elbow.
Step 7 - Connect the elbow to your mask.
Step 8 - To ensure your fit will work with your therapy: Switch on your device and position yourself as if you are to begin therapy (e.g. lying down in bed). Adjust your upper straps. Adjust your lower straps. Practice reseating your cushion – pull your mask away from your face, allow your cushion to reinflate, and then resettle the inflated cushion so the cushion membrane is sitting at the top of your nasal bridge.
Step 1 - Leak often occurs when your mask’s seal is compromised by movement, incorrect fitting or improper sizing.
Step 2 - To find a source of leak, run your finger along the edge of your mask’s seal. Once located, adjust your headgear.
Step 3 - Resolve any leak in the upper part of your mask by adjusting your upper straps.
Step 4 - Resolve lower mask leak with your lower straps.
Step 5 - If leak persists, reseal by briefly pulling your mask away from your face, letting your cushion reinflate, and resettling your mask cushion so the cushion membrane is sitting at the top of your nasal bridge.
Step 6 - If leak continues, you may need a different size mask. Try our sizing guide or speak to your healthcare provider about getting a different size mask.
Step 1 - Disassemble mask.
Step 2 - Rinse frame, elbow and cushion under running water. Clean each component with a soft brush.
Step 3 - Soak components in warm water with mild liquid detergent for up to ten minutes.
Step 4 - Shake components in water.
Step 5 - Brush moving parts of elbow and around vent holes.
Step 6 - Brush areas of frame where arms connect, and inside and outside the frame where the elbow connects.
Step 7 - Rinse components under running water.
Step 8 - Leave components to air dry out of direct sunlight – making sure to squeeze the arms of the frame to ensure excess water is removed.
Step 1 -Disassemble mask. EasyClick magnets can remain attached to headgear during cleaning.
Step 2 -Handwash headgear in warm water with mild liquid detergent.
Step 3 -Rinse headgear under running water. Inspect to ensure headgear is clean and detergent free. Wash and rinse again, if necessary.
Step 4 -Squeeze headgear to remove excess water.
Step 5 -Leave headgear to air dry out of direct sunlight.
Step 1 - Print you ResMed sizing guide.Remember to print to scale.
Step 2 - Trim your guide from the paper.
Step 3 - Using a mirror, hold your sizing guide to your face. The top of the guide should correspond with the highest part of tour nose bridge—just between your eyes.
Step 4 - The lower part of the guide indicates which mask size is best for your face. The size that aligns just below your lower lip will be your best fit.
Step 1 -Print your ResMed sizing guide.Remember to print to scale.
Step 2 -Trim your guide from the paper.
Step 3 -Using a mirror, hold your sizing guide to your face. The top of the guide should correspond with the highest part of tour nose bridge—just between your eyes.
Step 4 -The lower part of the guide indicates which mask size is best for your face. The size that aligns just below your lower lip will be your best fit.
Do you still have a question?