Designed to comfortably fit a broader range of patients, AirFit F20 features an entirely new cushion design. To make fitting your existing patients easier, we recommend using our sizing guide – included in each AirFit F20 box and specifically designed to help patients find the right fit.
You can also download a copy of sizing guide here.
The arrow of your sizing guide should align with the top of your patient’s nasal bridge. The segment of the guide corresponding with the crease below your patient’s lower lip will indicate their recommended fit. If a patient falls between sizes, we recommend selecting a larger option for greater comfort.
AirFit F20 has three cushion sizes - small, medium and large. The modular design of the AirFit F20 means each size cushion easily clicks into the same mask frame for easier setup.
Unclip EasyClick Magnets.
With your patient holding the mask towards their face, slide the headgear over the back of their head.
Reclip EasyClick Magnets.
Use the upper and lower headgear straps to help fine-tune their fit. Do not overtighten as this may cause discomfort.
This is also a good opportunity for your patient to reseat their cushion for comfort. When fitting is concluded, the cushion membrane should be fitting comfortably over your patient’s nasal bridge.
If your patient experiences leaks, or for additional information on mask maintenance and care, you can also visit the Supporting your patients section for more videos.
AirView is ResMed’s cloud-based patient management system that provides remote access to your patients’ therapy data. Patients using AirSense™ 10 or AirCurve™ 10 devices with built-in wireless connectivity can be remotely monitored.
* Wireless communication depends on network availability.
With the mask held against your face, pull the headgear over your head. Ensure the top of the mask cushion is sitting comfortably at the top of your nasal bridge. The bottom of your headgear should settle comfortably at the top of your neck - with the top of the headgear sitting on your crown.
Bring each of the magnets up to meet their corresponding clips on the frame.
Adjust the tabs on the upper headgear.
Adjust the tabs on the lower headgear.
With your device still switched off, connect your device tubing to your mask elbow.
Connect the elbow to your mask.
To ensure your fit will work with your therapy:
Switch on your device and position yourself as if you are to begin therapy (e.g. lying down in bed).
Adjust your upper straps.
Adjust your lower straps.
Practice reseating your cushion – pull your mask away from your face, allow your cushion to reinflate, and then resettle the inflated cushion so the cushion membrane is sitting at the top of your nasal bridge.
Managing leak with your AirFit F20
Leak often occurs when your mask’s seal is compromised by movement, incorrect fitting or improper sizing.
To find a source of leak, run your finger along the edge of your mask’s seal. Once located, adjust your headgear.
Resolve any leak in the upper part of your mask by adjusting your upper straps.
Resolve lower mask leak with your lower straps.
If leak persists, reseal by briefly pulling your mask away from your face, letting your cushion reinflate, and resettling your mask cushion so the cushion membrane is sitting at the top of your nasal bridge.
If leak continues, you may need a different size mask. Try our sizing guide or speak to your healthcare provider about getting a different size mask.
Cleaning your AirFit F20 (Daily)
Rinse frame, elbow and cushion under running water. Clean each component with a soft brush.
Soak components in warm water with mild liquid detergent for up to ten minutes.
Shake components in water.
Brush moving parts of elbow and around vent holes.
Brush areas of frame where arms connect, and inside and outside the frame where the elbow connects.
Rinse components under running water.
Leave components to air dry out of direct sunlight – making sure to squeeze the arms of the frame to ensure excess water is removed.
Cleaning your AirFit F20 (Weekly)
Disassemble mask. EasyClick magnets can remain attached to headgear during cleaning.
Handwash headgear in warm water with mild liquid detergent.
Rinse headgear under running water. Inspect to ensure headgear is clean and detergent free. Wash and rinse again, if necessary.
To offer your patients extra support, you can introduce them to myAir, ResMed’s online patient support program for users of AirSense™ 10 and AirCurve™ 10 devices with built-in wireless connectivity.
*Wireless communication depends on network availability.